Get Tandem Mass Spectrometry essential facts below. View Videos or join the Tandem Mass Spectrometry discussion. Add Tandem Mass Spectrometry to your PopFlock.com topic list for future reference or share this resource on social media.
The molecules of a given sample are ionized and the first spectrometer (designated MS1) separates these ions by their mass-to-charge ratio (often given as m/z or m/Q). Ions of a particular m/z-ratio coming from MS1 are selected and then made to split into smaller fragment ions, e.g. by collision-induced dissociation, ion-molecule reaction, or photodissociation. These fragments are then introduced into the second mass spectrometer (MS2), which in turn separates the fragments by their m/z-ratio and detects them. The fragmentation step makes it possible to identify and separate ions that have very similar m/z-ratios in regular mass spectrometers.
Tandem mass spectrometry includes triple quadrupole mass spectrometer (qqq), quad time of flight (Q-tof), and hybrid mass spectrometer
Triple quadrupole mass spectrometer (QQQ)
Tandem mass spectrometry that the first and third quadrupole work as a mass filter. When analytes pass the second quadrupole, the fragmentation proceeds through collision with gas. Usually used for the pharmaceutical industry.
Quadrupole time of flight (Q-tof)
Q-tof mass spectrometer combines TOF and quadrupole instruments, which cause high mass accuracy for product ions, accurate quantitation capability, and fragmentation experiment applicability. This is a method of mass spectrometry that ion fragmentation (m/z) ratio determined through a time of flight measurement.
hybrid mass spectrometer
hybrid mass spectrometer consists of more than two mass analyzers.
Schematic of tandem mass spectrometry
Multiple stages of mass analysis separation can be accomplished with individual mass spectrometer elements separated in space or using a single mass spectrometer with the MS steps separated in time. For tandem mass spectrometry in space, the different elements are often noted in a shorthand, giving the type of mass selector used.
Tandem in space
Triple quadrupole diagram; and example of tandem mass spectrometry in space.
An ion trap mass spectrometer is an example of a tandem mass spectrometry in time instrument.
By doing tandem mass spectrometry in time, the separation is accomplished with ions trapped in the same place, with multiple separation steps taking place over time. A quadrupole ion trap or Fourier transform ion cyclotron resonance (FTICR) instrument can be used for such an analysis. Trapping instruments can perform multiple steps of analysis, which is sometimes referred to as MSn (MS to the n). Often the number of steps, n, is not indicated, but occasionally the value is specified; for example MS3 indicates three stages of separation.
Tandem in time MS instruments do not use the modes described next, but typically collect all of the information from a precursor ion scan and a parent ion scan of the entire spectrum. Each instrumental configuration utilizes a unique mode of mass identification.
Tandem in space MS/MS modes
When tandem MS is performed with an in space design, the instrument must operate in one of a variety of modes. There are a number of different tandem MS/MS experimental setups and each mode has its own applications and provides different information. Tandem MS in space uses the coupling of two instrument components which measure the same mass spectrum range but with a controlled fractionation between them in space, while tandem MS in time involves the use of an ion trap.
There are four main scan experiments possible using MS/MS: precursor ion scan, product ion scan, neutral loss scan, and selected reaction monitoring.
For a precursor ion scan, the product ion is selected in the second mass analyzer, and the precursor masses are scanned in the first mass analyzer. Note that precursor ion is synonymous with parent ion and product ion with daughter ion; however the use of these anthropomorphic terms is discouraged.
In a product ion scan, a precursor ion is selected in the first stage, allowed to fragment and then all resultant masses are scanned in the second mass analyzer and detected in the detector that is positioned after the second mass analyzer. This experiment is commonly performed to identify transitions used for quantification by tandem MS.
In a neutral loss scan, the first mass analyzer scans all the masses. The second mass analyzer also scans, but at a set offset from the first mass analyzer. This offset corresponds to a neutral loss that is commonly observed for the class of compounds. In a constant-neutral-loss scan, all precursors that undergo the loss of a specified common neutral are monitored. To obtain this information, both mass analyzers are scanned simultaneously, but with a mass offset that correlates with the mass of the specified neutral. Similar to the precursor-ion scan, this technique is also useful in the selective identification of closely related class of compounds in a mixture.
In selected reaction monitoring, both mass analyzers are set to a selected mass. This mode is analogous to selected ion monitoring for MS experiments. A selective analysis mode, which can increase sensitivity.
Fragmentation of gas-phase ions is essential to tandem mass spectrometry and occurs between different stages of mass analysis. There are many methods used to fragment the ions and these can result in different types of fragmentation and thus different information about the structure and composition of the molecule.
Often, the ionization process is sufficiently violent to leave the resulting ions with sufficient internal energy to fragment within the mass spectrometer. If the product ions persist in their non-equilibrium state for a moderate amount of time before auto-dissociation this process is called metastable fragmentation. Nozzle-skimmer fragmentation refers to the purposeful induction of in-source fragmentation by increasing the nozzle-skimmer potential on usually electrospray based instruments. Although in-source fragmentation allows for fragmentation analysis, it is not technically tandem mass spectrometry unless metastable ions are mass analyzed or selected before auto-dissociation and a second stage of analysis is performed on the resulting fragments. In-source fragmentation is often used in addition to tandem mass spectrometry (with post-source fragmentation) to allow for two steps of fragmentation in a pseudo MS3-type of experiment.
Post-source fragmentation is most often what is being used in a tandem mass spectrometry experiment. Energy can also be added to the ions, which are usually already vibrationally excited, through post-source collisions with neutral atoms or molecules, the absorption of radiation, or the transfer or capture of an electron by a multiply charged ion. Collision-induced dissociation (CID), also called collisionally activated dissociation (CAD), involves the collision of an ion with a neutral atom or molecule in the gas phase and subsequent dissociation of the ion. For example, consider
where the ion AB+ collides with the neutral species M and subsequently breaks apart. The details of this process are described by collision theory. Due to different instrumental configuration, two main different types of CID are possible: (i) beam-type (in which precursor ions are fragmented on-the-flight) and (ii) ion trap-type (in which precursor ions are first trapped, and then fragmented).
A third and more recent type of CID fragmentation is higher-energy collisional dissociation (HCD). HCD is a CID technique specific to orbitrap mass spectrometers in which fragmentation takes place external to the ion trap, it happens in the HCD cell (in some instruments named "ion routing multipole"). HCD is a trap-type fragmentation that has been shown to have beam-type characteristics.
Electron capture and transfer methods
The energy released when an electron is transferred to or captured by a multiply charged ion can induce fragmentation.
ETD cleaves randomly along the peptide backbone (c and z ions) while side chains and modifications such as phosphorylation are left intact. The technique only works well for higher charge state ions (z>2), however relative to collision-induced dissociation (CID), ETD is advantageous for the fragmentation of longer peptides or even entire proteins. This makes the technique important for top-down proteomics. Much like ECD, ETD is effective for peptides with modifications such as phosphorylation.
Electron-transfer and higher-energy collision dissociation (EThcD) is a combination ETD and HCD where the peptide precursor is initially subjected to an ion/ion reaction with fluoranthene anions in a linear ion trap, which generates c- and z-ions. In the second step HCD all-ion fragmentation is applied to all ETD derived ions to generate b- and y- ions prior to final analysis in the orbitrap analyzer. This method employs dual fragmentation to generate ion- and thus data-rich MS/MS spectra for peptide sequencing and PTM localization.
Negative electron transfer dissociation
Fragmentation can also occur with a deprotonated species, in which an electron is transferred from the species to an cationic reagent in a negative electron transfer dissociation (NETD):
NETD is compatible with fragmenting peptide and proteins along the backbone at the C?-C bond. The resulting fragments are usually ao- and x-type product ions.
Electron-detachment dissociation (EDD) is a method for fragmenting anionic species in mass spectrometry. It serves as a negative counter mode to electron capture dissociation. Negatively charged ions are activated by irradiation with electrons of moderate kinetic energy. The result is ejection of electrons from the parent ionic molecule, which causes dissociation via recombination.
Charge transfer dissociation
Reaction between positively charged peptides and cationic reagents, also known as charge transfer dissociation (CTD), has recently been demonstrated as an alternative high-energy fragmentation pathway for low-charge state (1+ or 2+) peptides. The proposed mechanism of CTD using helium cations as the reagent is:
Initial reports are that CTD causes backbone C?-C bond cleavage of peptides and provides ao- and x-type product ions.
The energy required for dissociation can be added by photon absorption, resulting in ion photodissociation and represented by
where represents the photon absorbed by the ion. Ultraviolet lasers can be used, but can lead to excessive fragmentation of biomolecules.
With surface-induced dissociation (SID), the fragmentation is a result of the collision of an ion with a surface under high vacuum. Today, SID is used to fragment a wide range of ions. Years ago, it was only common to use SID on lower mass, singly charged species because ionization methods and mass analyzer technologies weren't advanced enough to properly form, transmit, or characterize ions of high m/z. Over time, self-assembled monolayer surfaces (SAMs) composed of CF3(CF2)10CH2CH2S on gold have been the most prominently used collision surfaces for SID in a tandem spectrometer. SAMs have acted as the most desirable collision targets due to their characteristically large effective masses for the collision of incoming ions. Additionally, these surfaces are composed of rigid fluorocarbon chains, which don't significantly dampen the energy of the projectile ions. The fluorocarbon chains are also beneficial because of their ability to resist facile electron transfer from the metal surface to the incoming ions. SID's ability to produce subcomplexes that remain stable and provide valuable information on connectivity is unmatched by any other dissociation technique. Since the complexes produced from SID are stable and retain distribution of charge on the fragment, this produces a unique, spectra which the complex centers around a narrower m/z distribution. The SID products and the energy at which they form are reflective of the strengths and topology of the complex. The unique dissociation patterns help discover the Quaternary structure of the complex. The symmetric charge distribution and dissociation dependence are unique to SID and make the spectra produced distinctive from any other dissociation technique.
The SID technique is also applicable to ion-mobility mass spectrometry (IM-MS). Three different methods for this technique include analyzing the characterization of topology, intersubunit connectivity, and the degree of unfolding for protein structure. Analysis of protein structure unfolding is the most commonly used application of the SID technique. For Ion-mobility mass spectrometry (IM-MS), SID is used for dissociation of the source activated precursors of three different types of protein complexes: C-reactive protein (CRP), transthyretin (TTR), and concanavalin A (Con A). This method is used to observe the unfolding degree for each of these complexes. For this observation, SID showed the precursor ions' structures that exist before the collision with the surface. IM-MS utilizes the SID as a direct measure of the conformation for each proteins' subunit.
Fourier-transform ion cyclotron resonance (FTICR) are able to provide ultrahigh resolution and high mass accuracy to instruments that take mass measurements. These features make FTICR mass spectrometers a useful tool for a wide variety of applications such as several dissociation experiments such as collision-induced dissociation (CID, electron transfer dissociation (ETD), and others. In addition, surface-induced dissociation has been implemented with this instrument for the study of fundamental peptide fragmentation. Specifically, SID has been applied to the study of energetics and the kinetics of gas-phase fragmentation within an ICR instrument. This approach has been used to understand the gas-phase fragmentation of protonated peptides, odd-electron peptide ions, non-covalent ligand-peptide complexes, and ligated metal clusters.
Quantitative proteomics is used to determine the relative or absolute amount of proteins in a sample. Several quantitative proteomics methods are based on tandem mass spectrometry. MS/MS has become a benchmark procedure for the structural elucidation of complex biomolecules.
One method commonly used for quantitative proteomics is isobaric tag labeling. Isobaric tag labeling enables simultaneous identification and quantification of proteins from multiple samples in a single analysis. To quantify proteins, peptides are labeled with chemical tags that have the same structure and nominal mass, but vary in the distribution of heavy isotopes in their structure. These tags, commonly referred to as tandem mass tags, are designed so that the mass tag is cleaved at a specific linker region upon higher-energy collisional-induced dissociation (HCD) during tandem mass spectrometry yielding reporter ions of different masses. Protein quantitation is accomplished by comparing the intensities of the reporter ions in the MS/MS spectra. Two commercially available isobaric tags are iTRAQ and TMT reagents.
Isobaric tags for relative and absolute quantitation (iTRAQ)
Isobaric labeling for tandem mass spectrometry: proteins are extracted from cells, digested, and labeled with tags of the same mass. When fragmented during MS/MS, the reporter ions show the relative amount of the peptides in the samples.
A tandem mass tag (TMT) is an isobaric mass tag chemical label used for protein quantification and identification. The tags contain four regions: mass reporter, cleavable linker, mass normalization, and protein reactive group. TMT reagents can be used to simultaneously analyze 2 to 11 different peptide samples prepared from cells, tissues or biological fluids. Three types of TMT reagents are available with different chemical reactivities: (1) a reactive NHS ester functional group for labeling primary amines (TMTduplex, TMTsixplex, TMT10plex plus TMT11-131C), (2) a reactive iodoacetyl functional group for labeling free sulfhydryls (iodoTMT) and (3) reactive alkoxyamine functional group for labeling of carbonyls (aminoxyTMT).
Chromatography trace (top) and tandem mass spectrum (bottom) of a peptide.
Tandem mass spectrometry can be used for protein sequencing. When intact proteins are introduced to a mass analyzer, this is called "top-down proteomics" and when proteins are digested into smaller peptides and subsequently introduced into the mass spectrometer, this is called "bottom-up proteomics". Shotgun proteomics is a variant of bottom up proteomics in which proteins in a mixture are digested prior to separation and tandem mass spectrometry.
Tandem mass spectrometry can produce a peptide sequence tag that can be used to identify a peptide in a protein database. A notation has been developed for indicating peptide fragments that arise from a tandem mass spectrum. Peptide fragment ions are indicated by a, b, or c if the charge is retained on the N-terminus and by x, y or z if the charge is maintained on the C-terminus. The subscript indicates the number of amino acid residues in the fragment. Superscripts are sometimes used to indicate neutral losses in addition to the backbone fragmentation, * for loss of ammonia and ° for loss of water. Although peptide backbone cleavage is the most useful for sequencing and peptide identification other fragment ions may be observed under high energy dissociation conditions. These include the side chain loss ions d, v, w and ammonium ions and additional sequence-specific fragment ions associated with particular amino acid residues.
Oligosaccharides may be sequenced using tandem mass spectrometry in a similar manner to peptide sequencing. Fragmentation generally occurs on either side of the glycosidic bond (b, c, y and z ions) but also under more energetic conditions through the sugar ring structure in a cross-ring cleavage (x ions). Again trailing subscripts are used to indicate position of the cleavage along the chain. For cross ring cleavage ions the nature of the cross ring cleavage is indicated by preceding superscripts.
Tandem mass spectrometry cannot be applied for single-cell analyses. it is insensitive to analyze such small amounts of a cell. Theses limitations are primarily due to a combination of inefficient ion production. Ion loses within the instruments due to chemical noise sources of solvents.
Tandem mass spectrometry will be a useful tool for protein characterization, nucleoprotein complexes, and other biological structures. However, some challenges left such as analyzing the characterization of the proteome quantitatively and qualitatively.
^Körner R, Wilm M, Morand K, Schubert M, Mann M (February 1996). "Nano electrospray combined with a quadrupole ion trap for the analysis of peptides and protein digests". Journal of the American Society for Mass Spectrometry. 7 (2): 150-6. doi:10.1016/1044-0305(95)00626-5. PMID24203235.
^ abOlsen JV, Macek B, Lange O, Makarov A, Horning S, Mann M (September 2007). "Higher-energy C-trap dissociation for peptide modification analysis". Nature Methods. 4 (9): 709-12. doi:10.1038/nmeth1060. PMID17721543.
^Senko MW, Remes PM, Canterbury JD, Mathur R, Song Q, Eliuk SM, Mullen C, Earley L, Hardman M, Blethrow JD, Bui H, Specht A, Lange O, Denisov E, Makarov A, Horning S, Zabrouskov V (December 2013). "Novel parallelized quadrupole/linear ion trap/Orbitrap tribrid mass spectrometer improving proteome coverage and peptide identification rates". Analytical Chemistry. 85 (24): 11710-4. doi:10.1021/ac403115c. PMID24251866.
^Nagaraj N, D'Souza RC, Cox J, Olsen JV, Mann M (December 2010). "Feasibility of large-scale phosphoproteomics with higher energy collisional dissociation fragmentation". Journal of Proteome Research. 9 (12): 6786-94. doi:10.1021/pr100637q. PMID20873877.
^Frese CK, Altelaar AF, van den Toorn H, Nolting D, Griep-Raming J, Heck AJ, Mohammed S (November 2012). "Toward full peptide sequence coverage by dual fragmentation combining electron-transfer and higher-energy collision dissociation tandem mass spectrometry". Analytical Chemistry. 84 (22): 9668-73. doi:10.1021/ac3025366. PMID23106539.
^Coon JJ, Shabanowitz J, Hunt DF, Syka JE (June 2005). "Electron transfer dissociation of peptide anions". Journal of the American Society for Mass Spectrometry. 16 (6): 880-2. doi:10.1016/j.jasms.2005.01.015. PMID15907703.
^Chingin K, Makarov A, Denisov E, Rebrov O, Zubarev RA (January 2014). "Fragmentation of positively-charged biological ions activated with a beam of high-energy cations". Analytical Chemistry. 86 (1): 372-9. doi:10.1021/ac403193k. PMID24236851.
^ abStiving, Alyssa; VanAernum, Zachary; Busch, Florian; Harvey, Sophie; Sarni, Samantha; Wysocki, Vicki (9 November 2018). "Surface-Induced Dissociation: An Effective Method for Characterization of Protein Quaternary Structure". Analytical Chemistry. 91: 190-191. doi:10.1021/acs.analchem.8b05071.
^Quintyn, Royston S.; Zhou, Mowei; Yan, Jing; Wysocki, Vicki H. (1 December 2015). "Surface-Induced Dissociation Mass Spectra as a Tool for Distinguishing Different Structural Forms of Gas-Phase Multimeric Protein Complexes". Analytical Chemistry. 87 (23): 11879-11886. doi:10.1021/acs.analchem.5b03441. ISSN0003-2700.
^Kaplan, Desmond A.; Hartmer, Ralf; Speir, J. Paul; Stoermer, Carsten; Gumerov, Dmitry; Easterling, Michael L.; Brekenfeld, Andreas; Kim, Taeman; Laukien, Frank; Park, Melvin A. (2008). "Electron transfer dissociation in the hexapole collision cell of a hybrid quadrupole-hexapole Fourier transform ion cyclotron resonance mass spectrometer". Rapid Communications in Mass Spectrometry. 22 (3): 271-278. Bibcode:2008RCMS...22..271K. doi:10.1002/rcm.3356. ISSN0951-4198. PMID18181247.
^Laskin, Julia (June 2015). "Surface-Induced Dissociation: A Unique Tool for Studying Energetics and Kinetics of the Gas-Phase Fragmentation of Large Ions". European Journal of Mass Spectrometry. 21 (3): 377-389. doi:10.1255/ejms.1358. ISSN1469-0667.
^Bantscheff M, Schirle M, Sweetman G, Rick J, Kuster B (October 2007). "Quantitative mass spectrometry in proteomics: a critical review". Analytical and Bioanalytical Chemistry. 389 (4): 1017-31. doi:10.1007/s00216-007-1486-6. PMID17668192.
^Ross PL, Huang YN, Marchese JN, Williamson B, Parker K, Hattan S, Khainovski N, Pillai S, Dey S, Daniels S, Purkayastha S, Juhasz P, Martin S, Bartlet-Jones M, He F, Jacobson A, Pappin DJ (December 2004). "Multiplexed protein quantitation in Saccharomyces cerevisiae using amine-reactive isobaric tagging reagents". Molecular & Cellular Proteomics. 3 (12): 1154-69. doi:10.1074/mcp.M400129-MCP200. PMID15385600.
^Zieske LR (2006). "A perspective on the use of iTRAQ reagent technology for protein complex and profiling studies". Journal of Experimental Botany. 57 (7): 1501-8. doi:10.1093/jxb/erj168. PMID16574745.
^Thompson A, Schäfer J, Kuhn K, Kienle S, Schwarz J, Schmidt G, Neumann T, Johnstone R, Mohammed AK, Hamon C (April 2003). "Tandem mass tags: a novel quantification strategy for comparative analysis of complex protein mixtures by MS/MS". Analytical Chemistry. 75 (8): 1895-904. doi:10.1021/ac0262560. PMID12713048.
^Shadforth I, Crowther D, Bessant C (November 2005). "Protein and peptide identification algorithms using MS for use in high-throughput, automated pipelines". Proteomics. 5 (16): 4082-95. doi:10.1002/pmic.200402091. PMID16196103.
^Roepstorff P, Fohlman J (November 1984). "Proposal for a common nomenclature for sequence ions in mass spectra of peptides". Biomedical Mass Spectrometry. 11 (11): 601. doi:10.1002/bms.1200111109. PMID6525415.
^Falick AM, Hines WM, Medzihradszky KF, Baldwin MA, Gibson BW (November 1993). "Low-mass ions produced from peptides by high-energy collision-induced dissociation in tandem mass spectrometry". Journal of the American Society for Mass Spectrometry. 4 (11): 882-93. doi:10.1016/1044-0305(93)87006-X. PMID24227532.
^Banoub JH, Newton RP, Esmans E, Ewing DF, Mackenzie G (May 2005). "Recent developments in mass spectrometry for the characterization of nucleosides, nucleotides, oligonucleotides, and nucleic acids". Chemical Reviews. 105 (5): 1869-915. doi:10.1021/cr030040w. PMID15884792.
^Tarini BA (August 2007). "The current revolution in newborn screening: new technology, old controversies". Archives of Pediatrics & Adolescent Medicine. 161 (8): 767-72. doi:10.1001/archpedi.161.8.767. PMID17679658.
^Chace DH, Kalas TA, Naylor EW (November 2003). "Use of tandem mass spectrometry for multianalyte screening of dried blood specimens from newborns". Clinical Chemistry. 49 (11): 1797-817. doi:10.1373/clinchem.2003.022178. PMID14578311.